An LCD Digital Thermometer for Convenient & Accurate Reads
The Apex Large Face Digital Thermometer enables users you to take a temperature quickly and accurately. Its large face combined with a color bar indicator and beeper makes it easy to read. The device is able to read temperatures in just five seconds and in either Celsius or Fahrenheit.
Features & Benefits
- Reads temp in 10 seconds
- Color bar indicator shows if fever is present
- Easy to read with large LCD screen
- Flexible tip prevents injury
- Reads temperature in Celsius or Fahrenheit
- Stores last measured temperature
Large LCD Display Makes it Easy to Read
The large LCD on this digital thermometer enables an easy read. It takes just five seconds to read the temperature and beeps when finished.
Color Bar Indicator Tells if Fever is Present
Worried about a fever? The device's color bar indicator shows red when a fever is present.
Flexible Tip Prevents Injury
Use the thermometer orally, rectally, or under the arm. It features a comfortable and safe flexible tip to prevent injury; perfect for kids.
Convenient, Hygienic Case Included
Keep the thermometer stored in its hygienic case when not in use.
How do I calibrate this thermometer. Item # : 70256 Apex Large Face LCD Fast Read Digital Thermometer
You can find callibration instructions in the user manual here: https://cdn.shopify.com/s/files/1/0240/6504/8681/files/APX_70256_INST_02_sm.pdf?v=1585065810
my apex thermometer reads several degrees low, how do I calibrate the thermometer?
Here are the directions for calibrating:
"The thermometer is initially calibrated at the time of manufacture. lf the thermometer is used according
to the use instruction, periodic readjustment is not required. However, we recommend checking
calibration every two years or whenever clinical accuracy of the thermometer is in question. Turn on the
thermometer and insert into the water bath which the temperature is steady. ASTM laboratory accuracy
requirements in the display range of 98.6 to 102.2°F (37.0 to 39.0°C) for electronic thermometers is
±0.2°F(±0.1°C). The above recommendations do not supersede the legal requirements. The user must
always comply with legal requirements for the control of the measurement, functionality, and accuracy
of the device which are required by the scope of relevant laws, directives or ordinances where the device
What size batteries does this take?
This particular thermometer uses a 1.5 V DC button battery (size LR41, SR41, UCC 392, or equivalent).